A technique called Fourier series harmonic analysis can separately quantify regular and irregular corneal astigmatism and detect subtle shape changes from corneal disorders or anterior segment surgery. Recently, a case-control study used the analysis on data from both a Placido ring–based corneal topographer and a 3D anterior segment optical coherence tomographer (OCT) to better understand dry eye disease. The two sets measured a significantly higher amount of corneal irregular astigmatism in dry eyes than in normal eyes.
Researchers obtained topographical data from 44 subjects with dry eye and 20 normal control subjects. The team then compared spherical, regular astigmatism and irregular astigmatism (asymmetry and higher-order irregularity) refractive error components of the cornea. Because asymmetric values and higher-order irregularities from the topographer were signiﬁcantly greater in dry eyes with moderate/severe central superﬁcial punctate keratopathy (SPK) than in those with no or mild SPK, the ﬁndings suggest a relationship between central SPK and the severity of corneal irregular astigmatism in eyes with dry eye disease. The differences between mild and moderate/severe dry eye were greater than those between normal and mild dry eye.
The study noted that the Placido-based topographer uses the reﬂection of ring images projected onto the air–tear ﬁlm interface, making it more capable of detecting tear ﬁlm–induced changes in corneal topography. Still, Placido-based topography may overestimate tear ﬁlm changes if digitizing errors occur because of remarkably distorted rings, particularly in moderate or severe dry eye. Researchers propose that OCT may be predictive of corneal shape after the tear ﬁlm condition in a dry eye has improved.
While the two devices’ measurements were statistically different, the magnitudes of these differences were not clinically relevant. Nevertheless, the researchers advise not to use the two interchangeably for corneal irregular astigmatism measurements. Yet when a greater amount of corneal irregular astigmatism is detected with a Placido-based topographer than with OCT, they advise to suspect dry eye.
|Koh S, Maeda N, Ogawa M, et al. Fourier analysis of corneal irregular astigmatism due to the corneal surface in dry eye. Eye Contact Lens. December 11, 2018. [Epub ahead of print].